FermCalc - Alcohol Content Calculators - Hydrometer Temperature Corrections
- Hydrometer SG Drop Calculator
- Introduction
- Input Field Definitions
- Output Field Definitions
- Calculation Details
- Berry Method
- Duncan & Acton Method
- Balling Method
- Cutaia, Reid, & Speers Method
- Estimation of Solids Content (True Brix)
- Hydrometer and Refractometer Calculator
- Introduction
- Input Field Definitions
- Output Field Definitions
- Calculation Details
- Rogerson & Symington Method
- Son et al. Method
- Roesener Method
- Barth & Race Method
- Boiling (Spirit Indication) Calculator
- Introduction
- Input Field Definitions
- Output Field Definitions
- Calculation Details
- Tabarié (Division) Method
- Blunt (Subtraction) Method
- Honneyman Method
- Hackbarth Method
- OIML Calculator
- Introduction
- Input Field Definitions
- Calculation Details
## Hydrometer Temperature Corrections
FermCalc allows hydrometer readings to be corrected for temperature in all three
of the alcohol content calculation methods. The corrections are functions of
both the alcohol and solids content of the liquids. The alcohol correction is
determined from the OIML formula (OIML, 1973). The solids correction is
determined from a fit of tables published in the AOAC The SG measured with a hydrometer is the ratio of the density of the liquid divided by the density of water at the hydrometer calibration temperature, or:
where
To correct the reading to a reference temperature of 20°C, we simply need to multiply the hydrometer reading by the ratio of the water density at the old calibration temperature to the water density at 20°C, or:
where
Water densities are calculated using the OIML formula assuming 0% alcohol. After correcting for the hydrometer calibration temperature, we can correct for the sample temperature. The coefficient of thermal expansion for aqueous ethanol-sucrose solutions, which determines the change in density for a given change in temperature, depends on both the ethanol and sucrose levels, but mainly on the ethanol level (Espejo and Armada, 2011). To correct the measured SG, FermCalc determines a composite correction factor by first calculating three density correction factors as follows:
where
The corrected SG
The densites for water and ethanol-water solutions are based on the OIML formula. The change in density of sucrose-water solutions with temperature is based on the following curve fit of the AOAC brix correction tables:
where:
The plot below compares the AOAC data with the calculated Brix corrections from equation (7) above.
The temperature corrections are valid in the temperature range from 0°C to 40°C, which is the overlapping range of validity between the OIML formula and the AOAC tables. Some of the alcohol content calculation methods below are based on SG values at 60°F instead of 20°C, in which cases the appropriate densities are substituted into the above equations. FermCalc only reports the SG values corrected to 20°C, so the actual SG values being used in the equations will be slightly different than the reported SG values in these cases. Back to top## Hydrometer SG Drop Calculator## IntroductionThe Hydrometer SG Drop Calculator estimates the alcohol content of a wine from two specific gravity (SG) measurements: one obtained prior to fermentation, and one obtained after fermentation is complete. Four different calculation methods are used to estimate alcohol content: The calculator will adjust the hydrometer readings for temperature if the measurement temperatures and the hydrometer calibration temperatures are provided. If the entered SG values are already corrected for temperature, or if temperature corrections are not desired, simply enter 20°C (68°F) in all of the temperature fields and no corrections will be made. This calculation is only valid for SG or density measurements obtained with a hydrometer or some other device for directly measuring SG or density. It does not work with refractometer readings because of the obscuration effect of alcohol on refractive index. The final SG measurement must be taken prior to any post-fermentation sweetening. If any sweetener is added during fermentation after the initial SG measurement is taken, the initial SG measurement must be adjusted accordingly. Dr. William Honneyman (1966) compared alcohol levels calculated using the SG drop method to the distillation data of Thorpe & Brown (1914) and found that it gives results "reasonably comparable with tests by distillation, provided graded factors suited to each drop in gravity are used". While he did not recommend a specific way to calculate the graded factors, he provided an important validation of the approach in general. Back to top## Input Field DefinitionsHydrometer SG Reading – The initial and final hydrometer SG readings. The initial reading should be taken prior to fermentation, and the final SG reading should be taken after fermentation is complete. Range: 0.77193 to 1.55454 SG Reading Temperature – The temperatures of the must and wine samples at the time of the initial and final SG readings. Range: 0°C (32°F) to 40°C (104°F) Calibration Temperature – The hydrometer calibration temperature(s) of the hydrometer(s) used to take the initial and final SG readings. Range: 0°C (32°F) to 40°C (104°F) ## Output Field DefinitionsCorrected SG (20°C/20°C) – The temperature-corrected initial and final SG values. Alcohol Content – The alcohol content of the wine calculated from the difference between the initial and final corrected SG values using the four calculation methods listed above. True Brix (Solids Content) – The total solids content of the wine calculated from the final corrected SG value and the average alcohol content of the four calculation methods. ## Calculation Details## Berry Method
This is the most commonly used SG drop method, and is described on pages 79-80 of
where
Ritchie Products Co. (2004) claims to have compared the results of equation (8) to the results of gas chromatography for a wide range of wines, and found that the results were within 0.3% vol/vol. This method has a temperature basis of 15.56°C (60°F). Back to top## Duncan & Acton Method
This method is described on pages 64-66 of
where
Combining equations (9) through (11) above yields the following equation:
This method has a temperature basis of 15.56°C (60°F). Back to top## Balling Method
The Balling method is normally used for beer but gives results that agree very
well with the other methods. The equations used in FermCalc were taken from Michael
Hall's article "Brew by the Numbers" in the Summer 1995 issue of
where
The alcohol content (% by weight) is then calculated as:
where The result of equation (15) is then converted to % alcohol by volume as described here. This method has a temperature basis of 17.5°C (63.5°F). Back to top## Cutaia, Reid, & Speers MethodCutaia, Reid & Spears (2009) analyzed data from 532 beers to develop equation (12) below relating alcohol content to the initial and final specific gravities.
The alcohol content of the beers ranged from 3% to 7% by weight (approx. 3.8% to 8.7% by vol.). As with the Balling equation above, the specific gravities for the analyzed beers were expressed in degrees Plato, which is assumed by FermCalc to be the same as degrees Brix. The result of equation (16) is then converted to % alcohol by volume as described here. This method has a temperature basis of 20°C (68°F). Back to top## Estimation of Solids Content (True Brix)
After we know the alcohol content, we can estimate true Brix, which represents
solids content in % by weight, by using the model developed by James Hackbarth
(2011), which is described below. This is done by treating
the specific gravity ## Hydrometer & Refractometer Calculator## IntroductionThe Hydrometer and Refractometer Calculator estimates the alcohol content of a finished wine from a refractometer reading and a hydrometer specific gravity (SG) reading. Four different calculation methods are used to estimate alcohol content: All of the methods are designed to be used after fermentation, but they should be able to yield reasonable estimates of alcohol content during fermentation as long as there is enough alcohol to affect the measurements and the sample is degassed enough that the measurements are not affected by dissolved CO2. ## Input Field DefinitionsRefractometer Reading – The refractometer reading for the wine after fermentation. Range: 0° to 100°Brix Hydrometer SG Reading – The hydrometer SG reading for the wine sample. Range: 0.77193 to 1.55454 SG Reading Temperature – The temperature of the wine sample at the time of the SG reading. Range: 0°C (32°F) to 40°C (104°F) Calibration Temperature – The hydrometer calibration temperature of the hydrometer used to take SG reading. Range: 0°C (32°F) to 40°C (104°F) ## Output Field DefinitionsCorrected SG (20°C/20°C) – The temperature-corrected SG value. Alcohol Content – The alcohol content of the wine calculated using the four calculation methods listed above. True Brix – The total solids content of the wine calculated using the four calculation methods listed above. Back to top## Calculation Details## Rogerson & Symington MethodRogerson & Symington (2006) developed a method to estimate alcohol content and residual solids (true Brix) from based on refractometer and hydrometer readings on 35 port wines. In the words of the authors, "It is not applicable for the analysis of dry wines, whether fortified or not, which contain insufficient soluble solids for Baumé determination by hydrometer, and is yet to be evaluated for sweet table wines, such as sauternes." However it is included in FermCalc because many home winemakers seem to find it useful for monitoring fermentation progress and calculating alcohol content.
FermCalc first converts the hydrometer reading
where Alcohol content is then calculated as:
where
True Brix,
This method has a temperature basis of 20°C (68°F). Back to top## Son et al. MethodH. S. Son et al. (2009) developed the following six empirical equations based on refractometer, hydrometer, and alcohol content measurements on 30 wines before and during fermentation.
where
Equations (24) and (25) allow calculation of alcohol content and true Brix directly
from hydrometer and refractometer readings. However, I found these equations to
be inaccurate, yielding estimates of alcohol content that appear too high in the
lower-alcohol range and too low in the upper range. Instead of using equations
(24) and (25), I developed alternative equations from equations (20) through (23)
that appear much more accurate. Combining equations (22) and (23) to eliminate
Combining equations (20) and (21) to eliminate
Equations (26) and (27) are used by FermCalc to calculate alcohol content and true Brix. This method has a temperature basis of 20°C (68°F). Back to top## Roesener MethodThis method was published online by Werner Roesener (2001) and is very popular among home winemakers, but there is no documentation regarding the derivation of the equations. My testing indicates that it yields results that are very similar to the other methods. Simplifying the original equations we get:
where
This method has a temperature basis of 15.56°C (60°F). Back to top## Barth & Race MethodThis method was originally developed by Georg Barth (1905) in Germany for analyzing beers. The original equations are:
Where
FermCalc uses equations (33) and (34) to calculate alcohol content and true Brix because they were intended for higher alcohol concentrations and might be more applicable for winemaking calculations. The result of equation (32) is converted to % alcohol by volume as described here. This method has a temperature basis of 15.56°C (60°F). Back to top## Boiling (Spirit-Indication) Calculator## IntroductionThe Boiling (Spirit Indication) Calculator estimates the alcohol content of a wine from two specific gravity (SG) measurements: one obtained on a sample of the finished wine, and one obtained on a sample of the wine in which the alcohol has been boiled off and distilled water has been added to restore it to its pre-boiled volume. Four different calculation methods are used to estimate alcohol content: The calculator will adjust the hydrometer readings for temperature if the measurement temperatures and the hydrometer calibration temperatures are provided. If the entered SG values are already corrected for temperature, or if temperature corrections are not desired, simply enter 20°C (68°F) in all of the temperature fields and no corrections will be made. ## Input Field DefinitionsHydrometer SG Reading – The initial and final hydrometer SG readings. The initial reading should be taken prior to boiling, and the final SG reading should be taken after boiling off the alcohol and restoring the sample to its original volume with distilled water. Range: 0.77193 to 1.55454 SG Reading Temperature – The temperatures of the samples at the time of the initial and final SG readings. Range: 0°C (32°F) to 40°C (104°F) Calibration Temperature – The hydrometer calibration temperature(s) of the hydrometer(s) used to take the initial and final SG readings. Range: 0°C (32°F) to 40°C (104°F) ## Output Field DefinitionsCorrected SG (20°C/20°C) – The temperature-corrected initial and final SG values. Alcohol Content – The alcohol content of the wine calculated from the difference between the initial and final corrected SG values using the four calculation methods listed above. True Brix (Solids Content) – The total solids content of the wine calculated from the initial and final corrected SG values. ## Calculation DetailsThis method was first proposed by M. E. Tabarié in 1830 as a simplified alternative to the distillation procedure. It is based on the principle that alcohol causes the same depression in SG in wine as it does in pure water. The method involves evaporating (boiling off) a portion of the wine sample until all of the alcohol is evaporated, and then replacing the evaporated volume with distilled water. The difference between the specific gravities of the wine and the volume-corrected residue are then used to estimate the SG of the distillate, which represents the SG of a pure water/ethanol mixture, from which the alcohol content can be estimated. The experimental procedure is summarized below. - Measure the SG (
*sg*) of the wine to be tested._{w} - Take a sample of about 250-500 mL (1-2 cups) of the wine and boil the sample down to approximately half of its original volume to drive off all of the alcohol.
- Allow the boiled residue to cool to room temperature.
- Add distilled water to the residue until the total volume is restored to the original sample volume.
- Measure the SG of this volume-corrected residue
*sg*, which will be greater than_{r}*sg*because the alcohol has been replaced by water._{w}
It is recommended that a narrow-range hydrometer be used for the SG measurements since small errors in these measurements can result in large errors in the results. In addition to estimating the alcohol content, we can also calculate the solids content (true Brix) of the wine from the SG of the volume-corrected residue. First we just need to convert the residue SG measurement to a Brix value by using the Brix conversion equation. This conversion yields the solids content in % by weight of the residue. We then need to convert this value to the solids content in % by weight of the wine by multiplying by the ratio of specific gravities, or:
where
The four methods that FermCalc uses to calculate the alcohol content from the SG measurements are described below. Back to top## Tabarié (Division) MethodTabarié originally proposed estimating the SG of the distillate from the ratio of the wine and residue specific gravities, or:
where
FermCalc determines the alcohol content in % by volume from This method has a temperature basis of 20°C (68°F). Back to top## Blunt (Subtraction) Method
T. P. Blunt (1891) suggested that the Tabarié division formula always underestimates
alcohol content, and suggested calculating sg and
_{r}sg instead of the ratio, or.
_{w}
FermCalc uses the OIML formula to estimate alcohol content in % by volume from the results of equation (37).
S. Harvey (1892) presented experimental results suggesting that Blunt's formula
is more accurate than Tabarié's. However, a few pages later in the same issue
of This method has a temperature basis of 20°C (68°F). Back to top## Honneyman Method
This method is described on pages 124-126 of
I obtained a copy of Dr. Honneyman's book and was able to determine that his table is based on the alcoholometric tables of Thorpe (1915), which have a temperature basis of 60°F. With this information I was able to extend his table to higher alcohol levels and increase the resolution. Now with the appropriate temperature correction the Honneyman method gives results that are nearly identical to the other boiling methods. There are very small differences because the temperature corrections are not exact and because the basis of the Thorpe tables is different from that of the OIML equation. This method has a temperature basis of 15.56°C (60°F). Back to top## Hackbarth MethodJames Hackbarth (2009) showed that the Tabarié approach works reasonably well for dry wines and low-extract beers, but is inaccurate for beverages with higher alcohol and sugar concentrations due to solute-solute interactions that take place at the higher concentrations, an idea that was first proposed by Leonard (1897). Based on extensive laboratory experimentation and detailed analysis of the results, Hackbarth (2011) developed a new model for estimating the SG of a mixture from its sucrose (extract) and alcohol concentrations. Since the spirit-indication procedure gives us measurements of the wine SG and the extract concentration (true Brix), we can treat these as known quantities and use the Hackbarth model to solve for the alcohol content. The Hackbarth model utilizes the following equations.
where
FermCalc solves equations (38) through (44) iteratively using the alcohol content
calculated from the Blunt model as the initial estimate. The iteration loop is
repeated until the difference between the value of ^{-8}.This method has a temperature basis of 20°C (68°F). Back to top## OIML Calculator## Introduction
The OIML Calculator solves the general formula for calculating the densities of
mixtures of ethanol and water found in
## Input Field Definitions
Density – The density of the
ethanol/water mixture. This field can be either an input field
or an output field. If it is used as an input field, the alcohol
content is calculated from the entered values of density and
temperature. If alcohol content is entered, density is
calculated. Range: 771.93 to 1000 kg/m Temperature – The temperature at which to calculate the density or alcohol content for the ethanol/water mixture. If the temperature is changed, either the density or alcohol content will be recalculated, depending on which one was entered last. Range: -20°C (-4°F) to 40°C (104°F) Alcohol Content – The alcohol content of the ethanol/water mixture. This field can be either an input field or an output field. If it is used as an input field, the density is calculated from the entered values of alcohol content and temperature. If density is entered, alcohol content is calculated. Range: 0% to 100% vol/vol ## Calculation Details
The OIML Calculator uses the general formula for calculating the densities of
mixtures of ethanol and water found in
where
Given an alcohol content and a temperature, equation (45) can be solved directly for the density of the mixture. Given a density and a temperature, FermCalc calculates the associated alcohol content using an iterative technique.
© 2007-2016 Steve Gross |